Issue 
A&A
Volume 598, February 2017



Article Number  A25  
Number of page(s)  22  
Section  Cosmology (including clusters of galaxies)  
DOI  https://doi.org/10.1051/00046361/201629626  
Published online  26 January 2017 
QuickPol: Fast calculation of effective beam matrices for CMB polarization
^{1} Sorbonne Universités, UPMC Univ. Paris 6 & CNRS (UMR 7095): Institut d’Astrophysique de Paris, 98bis boulevard Arago, 75014 Paris, France
email: hivon@iap.fr
^{2} Institut de Planétologie et d’Astrophysique de Grenoble, Université Grenoble Alpes, CNRS (UMR 5274), 38000 Grenoble, France
^{3} Institut d’Astrophysique Spatiale, CNRS (UMR 8617) Université ParisSud 11, Bâtiment 121, 91405 Orsay, France
Received: 31 August 2016
Accepted: 2 October 2016
Current and planned observations of the cosmic microwave background (CMB) polarization anisotropies, with their ever increasing number of detectors, have reached a potential accuracy that requires a very demanding control of systematic effects. While some of these systematics can be reduced in the design of the instruments, others will have to be modeled and hopefully accounted for or corrected a posteriori. We propose QuickPol, a quick and accurate calculation of the full effective beam transfer function and of temperature to polarization leakage at the power spectra level, as induced by beam imperfections and mismatches between detector optical and electronic responses. All the observation details such as exact scanning strategy, imperfect polarization measurements, and flagged samples are accounted for. Our results are validated on Planck high frequency instrument (HFI) simulations. We show how the pipeline can be used to propagate instrumental uncertainties up to the final science products, and could be applied to experiments with rotating halfwave plates.
Key words: cosmic background radiation / cosmology: observations / polarization / methods: analytical
© ESO, 2017
1. Introduction
We are now entering an era of precise measurements of the cosmic microwave background (CMB) polarization, with potentially enough sensitivity to detect or even characterize the primordial tensorial B modes, the smoking gun of inflation (e.g., Zaldarriaga & Seljak 1997, and references therein). This raises expectations about the control and the correction of contaminations by astrophysical foregrounds, observational features, and instrumental imperfections. As it has in the past, progress will come from the synergy between instrumentation and data analysis. Improvements in instrumentation call for improved precision in final results, which are made possible by improved algorithms and the ability to deal with more and more massive data sets. In turn, expertise gained in data processing allows for better simulations that lead to new instrument designs and better suited observations. An example of such joint developments is the study of the impact of optics and electronicsrelated imperfections on the measured CMB temperature and polarization angular power spectra and their statistical isotropy. Systematic effects such as beam noncircularity, response mismatch in dual polarization measurements and scanning strategy imperfections, as well as how they can be mitigated, have been extensively studied in the preparation of forthcoming instruments (including, but not limited to Souradeep & Ratra 2001; Fosalba et al. 2002; Hu et al. 2003; Mitra et al. 2004, 2009; O’Dea et al. 2007; Rosset et al. 2007; Shimon et al. 2008; Miller et al. 2009a,b; Hanson et al. 2010; Leahy et al. 2010; Rosset et al. 2010; Ramamonjisoa et al. 2013; Rathaus & Kovetz 2014; Wallis et al. 2014; Pant et al. 2016), and during the analysis of data collected by WMAP^{1} (Smith et al. 2007; Hinshaw et al. 2007; Page et al. 2007) or Planck^{2} (Planck Collaboration VII 2014; Planck Collaboration XVII 2014; Planck Collaboration XI 2016) satellite missions.
At the same time, several deconvolution algorithms and codes have been proposed to clean up the CMB maps from such beamrelated effects prior to the computation of the power spectra, like PreBeam (ArmitageCaplan & Wandelt 2009), ArtDeco (Keihänen & Reinecke 2012), and in Bennett et al. (2013) and Wallis et al. (2015), or during their production (Keihänen et al. 2016).
Finally, in a related effort, the FEBeCoP pipeline, described in Mitra et al. (2011) and used in Planck data analysis (Planck Collaboration IV 2014; Planck Collaboration VII 2014), can be seen as a convolution facility, by providing, at arbitrary locations on the sky, the effective beam maps and point spread functions of a detector set, which, in turn, can be used for a MonteCarlo based description of the effective beam window functions for a given sky model.
In this paper, we introduce the QuickPol pipeline, an extension to polarization of the Quickbeam algorithm used in Planck Collaboration VII (2014). It allows a quick and accurate computation of the leakage and crosstalk between the various temperature and polarization power spectra (TT, EE, BB, TE, etc.) taking into account the exact scanning, sample flags, relative weights, and scanning beams of the considered set(s) of detectors. The end results are effective beam matrices describing, for each multipole ℓ, the mixing of the various spectra, independently of the actual value of the spectra. As we shall see, the impact of changing any timeindependent feature of the instrument, such as its beam maps, relative gain calibrations, detector orientations, and polarization efficiencies can be propagated within seconds to the final beam matrices products, allowing extremely fast MonteCarlo exploration of the experimental features. QuickPol is thus a powerful tool for both real data analysis and forthcoming experiments, simulations and design.
The paper is organized as follows. The mathematical formalism is exposed in Sect. 2 and analytical results are given in Sect. 3. The numerical implementation is detailed in Sect. 4 and compared to the results of Planck simulations in Sect. 5. Section 6 shows the propagation of instrumental uncertainties. We discuss briefly the case of rotating halfwave plates in Sect. 7 and conclude in Sect. 8.
2. Formalism
2.1. Data stream of a polarized detector
As usual in the study of polarization measurement, we will use Jones’ formalism to study the evolution of the electric component of an electromagnetic radiation in the optical system. Let us consider a quasi monochromatic^{3} radiation propagating along the z axis, and hitting the optical system at a position . The incoming electric field will be turned into e′(r) = J(r).e(r), where J(r) is the 2 × 2 complex Jones matrix of the system.
A rotation of the optical system by α around the z axis can be seen as a rotation of both the orientation and location of the incoming radiation by −α in the detector reference frame, and the same input radiation is now received as (1)with and the † sign representing the adjoint operation, which for a real rotation matrix, simply amounts to the matrix transpose. The measured signal is (4)with (5)We now introduce the Stokes parameters of the input signal (dropping the dependence on r) (6)and of the (unrotated) instrument response (7)to obtain(8)With the rotated instrument response: Following Rosset et al. (2010), we can specify the instrument as being a beam forming optics, followed by an imperfect polarimeter in the direction x, with 0 ≤ η ≤ 1, and having an overall optical efficiency 0 ≤ τ ≤ 1: (10)with (11)for a = x,y. The Stokes parameters of the instrument are then for .
If the beam is assumed to be perfectly copolarized, that is, it does not alter at all the polarization of the incoming radiation, with b_{xy} = b_{yx} = 0 and b_{xx} = b_{yy}, then , , and , , , Eqs. ((8), (9)) become (12)where (13)is the polar efficiency, such that 0 ≤ ρ ≤ 1 with ρ = 1 for a perfect polarimeter and ρ = 0 for a detector only sensitive to intensity. In the case of Planck high frequency instrument (HFI), Rosset et al. (2010) showed the measured polarization efficiencies to differ by Δρ′ = 1% to 16% from their ideal values, with an absolute statistical uncertainty generally below 1%. The particular case of copolarized beams is important because in most experimental setups, such as Planck, the beam response calibration is done on astronomical or artificial far field sources. Well known, compact, and polarized sources are generally not available to measure and and only the intensity beam response is measured. In the absence of reliable physical optics modeling of the beam response, one therefore has to assume and to be perfectly copolarized.
So far, we have only considered the optical beam response. We should also take into account the scanning beam, which is the convolution of the optical beam with the finite time response of the instrument (or its imperfect correction) as it moves around the sky, as described in Planck Collaboration VII (2014) and Planck Collaboration VII (2016). These time related effects can be a major source of elongation of the scanning beams, and can increase the beam mismatch among sibling detectors. If one assumes the motion of the detectors on the sky to be nearly uniform, as was the case for Planck, then optical beams can readily be replaced by scanning beams in the QuickPol formalism.
2.2. Spherical harmonics analysis
We now define the tools that are required to extend the above results to the full celestial sphere. The temperature T is a scalar quantity, while the linear polarization Q ± iU is of spin ± 2, and the circular polarization V is generally assumed to vanish. They can be written as linear combinations of spherical harmonics (SH): although one usually prefers the scalar and fixed parity E and B components (16)such that for X = T,E,B. In other terms (17)with (18)The sign convention used in Eq. (16) is consistent with Zaldarriaga & Seljak (1997) and the HEALPix^{4} library (Górski et al. 2005).
The response of a beam centered on the North pole can also be decomposed in SH coefficients while the coefficients of a rotated beam can be computed by noting that under a rotation of angle α around the direction r, the SH of spin s transform as (21)The elements of Wigner rotation matrices D are related to the SH via (Challinor et al. 2000) (22)with .
If the beam is assumed to be copolarized and coupled with a perfect polarimeter rotated by an angle γ, such that in Cartesian coordinates (or in (θ,φ) polar coordinates), simple relations between b_{ℓm} and can be established. For a Gaussian circular beam of full width half maximum (FWHM) and of throughput Challinor et al. (2000) found The factor c_{2} = e^{2σ2} in Eq. (23b) is such that c_{2}−1 < 1.1 × 10^{4} for θ_{FWHM} ≤ 1° and c_{2}−1 < 3.1 × 10^{6} for θ_{FWHM} ≤ 10′, and will be assumed to be c_{2} = 1 from now on. For a slightly elliptical Gaussian beam, Fosalba et al. (2002) found (24)while we show in Appendix G that Eq. (24) is true for arbitrarily shaped copolarized beams. This result can also be obtained by noting that an arbitrary beam is the sum of Gaussian circular beams with different FWHM and center (Tristram et al. 2004), each of them obeying Eq. (23b).
The detector associated to a beam is an imperfect polarimeter with a polarization efficiency ρ′ and the overall polarized response of the detector, in a referential aligned with its direction of polarization (the socalled Pxx coordinates in Planck parlance), reads (25)so that (26)We introduced ρ′ to distinguish it from the ρ value used in the mapmaking, as described below.
2.3. Map making equation
A polarized detector pointing, at time t, in the direction r_{t} on the sky, and being sensitive to the polarization with angle α_{t} with respect to the local meridian, measures (27)The factor 1/2 present in Eq. (8) is assumed to be absorbed in the gain calibration, performed on large scale temperature fluctuations, such as the CMB solar dipole (Planck Collaboration VIII 2014), and we assumed the circular polarization V to vanish. With the definitions introduced in Sect. 2.2, this becomes (28)The mapmaking formalism is set ignoring the beam effects, assuming a perfectly copolarized detector and an instrumental noise n (Tristram et al. 2011, and references therein), so that, for a detector j, Eq. (12) becomes (29)where the leading prefactors are here again absorbed in the gain calibration. Let us rewrite it as (30)with (Shimon et al. 2008) and P = Q + iU. Assuming the noise to be uncorrelated between detectors, with covariance matrix for detector j, the generalized least square solution of Eq. (29) for a set of detectors is (33)Let us now replace the ideal data stream (Eq. (29)) with the one obtained for arbitrary beams (Eq. (27)) and further assume that the noise is white and stationary with variance , so that . Let us also introduce the binary flag f_{j,t} used to reject individual time samples from the mapmaking process; Eq. (33) then becomes We have assumed here the pixels to be infinitely small, so that, starting with Eq. (28), the location of all samples in a pixel coincides with the pixel center. The effect of the pixel’s finite size and the socalled subpixel effects will be considered in Sect. 3.5.
2.4. Measured power spectra
To compute the crosspower spectrum of any two spin v_{1} and v_{2} maps, we first project each polarized component v of on the appropriate spin weighted sets of spherical harmonics, (36)and average these terms according to where Eq. (C.6) was used. The detailed derivation of this relation and its associated terms is given in Appendix A. Suffice it to say here that k_{u} terms are either 1 or 1/2, terms are inverse noiseweighted beam multipoles, and terms are effective weights describing the scanning and depending on the direction of polarization, hit redundancy (both from sky coverage and flagged samples), and noise level of detector j.
Equation (38) is therefore a generalization to noncircular beams of the pseudopower spectra measured on a masked or weighted map (Hivon et al. 2002; Hansen & Górski 2003), and extends to polarization the Quickbeam noncircular beam formalism used in the data analysis conducted by Planck Collaboration VII (2014). It also formally agrees with Hu et al. (2003)’s results on the impact of systematic effects on the polarization power spectra, with the functions absorbing the systematic effect parameters relative to detector j. In the next sections, we present the numerical results implied by this result and compare them on fullfledged PlanckHFI simulations.
3. Results
We now apply the QuickPol formalism to configurations representative of current or forthcoming CMB experiments, and to a couple of idealized test cases for which the expected result is already known, as a sanity check. The effect of the finite pixel size is also studied.
Fig. 1
Orientation of polarization measurements in Planck. The two left panels show, for an actual Planck detector, the maps of ⟨cos2α⟩ and ⟨sin2α⟩ respectively, where α is the direction of the polarizer with respect to the local Galactic meridian, which contributes to the spin 2 term defined in Eq. (A.3). The right panel shows the power spectrum of , multiplied by ℓ(ℓ + 1)/2π. 
Fig. 2
Same as Fig. 1 for an hypothetical detector of a LiteBIRDlike mission, except for the right panel plot which has a different yrange. 
3.1. A note about scanning strategies
To begin with, let us consider the scanning strategy of Planck and of another satellite mission optimized for the measurement of CMB polarization.
Figure 1 illustrates the orientation of the polarization measurements achieved in Planck. It shows, for an actual Planck detector, the maps of ⟨cos2α⟩ and ⟨sin2α⟩ respectively, where α is the direction of the polarizer with respect to the local Galactic meridian. These quantities contribute to the spin 2 term defined in Eq. (A.3). The large amplitude of these two maps is consistent with the fact that for a given detector, the orientation of the polarization measurements is mostly α and −α, as expected when detectors move on almost great circles with very little precession. Another striking feature is the relative smoothness of the maps, which translate into the power spectrum of peaking at low ℓ values.
Fig. 3
Effective beam window matrix introduced in Eq. (41) and detailed in Eq. (E.8a), for the crossspectra of two simulated Planck maps discussed in Sect. 5. Left panel: raw elements of , showing for each ℓ how the measured XY map angular power spectrum is impacted by the input TT spectrum, because of the observation of the sky with the beams. Right panel: blownup ratio of the nondiagonal elements to the diagonal ones: 100 W^{XY,TT}_{ℓ}/W^{TT,TT}_{ℓ} 
Figure 2 shows the same information for an hypothetical LiteBIRD^{5} like detector (but without halfwave plate modulation) in which we assumed the detector to cover a circle of 45° in radius in one minute, with its spin axis precessing with a period of four days at 50° from the antisun direction. As expected for such a scanning strategy, the values of α are pretty uniformly distributed over the range [0,2π], which translates into a low amplitude of the ⟨cos2α⟩ and ⟨sin2α⟩ maps. Even if those maps do not look as smooth as those of Planck, their power spectra peak at fairly low multipole values.
3.2. Arbitrary beams, smooth scanning case
If one assumes that ω_{s}(p) and vary slowly across the sky, as we just saw in the case of Planck and LiteBIRD – and probably a wider class of orbital and suborbital missions – then is dominated by low ℓ′ values and one expects ℓ ≃ ℓ′′ because of the triangle relation imposed by the 3J symbols (see Appendix C). If one further assumes C_{ℓ} and b_{ℓ} to vary slowly in ℓ, then Eqs. (C.5) and (C.9) can be used to impose s_{1} + v_{1} = s_{2} + v_{2} = s in Eq. (38) and provide (39)with As derived in Appendix E.1, Eq. (39) reduces to a mixing equation relating the observed crosspower spectra to the true ones: (41)with X,Y,X′,Y′ ∈ { T,E,B }.
In the smooth scanning case representative of past and forthcoming satellite missions, the effect of observing the sky with nonideal beams is therefore to couple the temperature and polarization power spectra at the same multipole ℓ through an extended beam window matrix , as illustrated in Fig. 3.
3.3. Arbitrary scanning, circular identical beams
If the scanning beams are now assumed to all be circular and identical, the measured will not depend on the details of the scanning strategy, orientation of the detectors, or relative weights of the detectors. We are indeed exactly in the ideal hypotheses of the map making formalism (Eq. (29)) and get the well known and simple result that the effect of the beam can be factored out.
If one considers detectors with identical circular copolarized beams, and whose actual polarization efficiency was used during the map making: , such that (42)then Eqs. (39) and (40b) feature terms like , which when written in a matrix form, verify the equality (43)according to Eq. (B.9). The measured power spectra are then (44)and for the Gaussian circular beam introduced in Eq. (23a). Obviously, these very simple results assume that the whole sky is observed. If not, the cutsky induced ℓ−ℓ and E−B coupling effects mentioned at the end of Sect. 2.4 have to be accounted for, as described, for example, in Chon et al. (2004), Mitra et al. (2009), Grain et al. (2009), and references therein.
3.4. Arbitrary beams, ideal scanning
Let us now consider the case of an ideal scanning of the sky, for which in any pixel p, the number of valid (unflagged) samples is the same for all detectors h_{j}(p) = h(p), and each detector j covers uniformly all possible orientations within that pixel along the duration of the mission. This constitutes the ideal limit aimed at by the scanning strategy illustrated in Fig. 2. The assumption of smooth scanning is then perfectly valid, and details of the calculations can be found in Appendix E.2. We find for instance that the matrix describing how the measured temperature and polarization power spectra are affected by the input TT spectrum reads (45)with the normalization factors (46)This confirms that in this ideal case, as expected and discussed previously (e.g., Wallis et al. 2014, and references therein), the leakage from temperature to polarization (Eq. (45)) is driven by the beam ellipticity ( terms) which has the same spin ± 2 as polarization. One also sees that the contamination of the E and B spectra by T are swapped (e.g., ) when the beams are rotated with respect to the polarimeter direction by 45° ( ), as shown in Shimon et al. (2008).
3.5. Finite pixel size and subpixel effects
As shown in Planck Collaboration VII (2014), in the case of temperature fluctuations, the effect of the finite pixel size is twofold. First, in each pixel, the distance between the nominal pixel center and the center of mass of the observations couples to the local gradient of the Stokes parameters to induce noise terms. Second, there is a smearing effect due to the integration of the signal over the surface of the pixel. Equation (41) then becomes (47)with , and the squared displacement averaged over the hits in the pixels and over the set of considered pixels. As shown in Appendix F, the additive noise term, sourced by the temperature gradient within the pixel, affects both temperature and polarization measurements, with and , while the other spectra are much less affected, that is, . The sign of this noise term is arbitrary and can be negative when crosscorrelating maps with a different sampling of the pixels.
4. Numerical implementation
Numerical implementations of this formalism are performed in three steps, assuming that the individual beam is already computed for 0 ≤ s ≤ s_{max} + 4 and 0 ≤ ℓ ≤ ℓ_{max}:
 1.
For each involved detector j, and for 0 ≤ s ≤ s_{max}, one computes the sth complex moment of its direction of polarization in pixel p: defined in Eq. (A.3). Since this requires processing the whole scanning data stream, this step can be time consuming. However it has to be computed only once for all cases, independently of the choices made elsewhere on the beam models, calibrations, noise weighting, and other factors. As we shall see below, it may not even be necessary to compute it, or store it, for every sky pixel.
 2.
The computed above are weighted with the assumed inverse noise variance weights w_{j} and polar efficiencies ρ_{j} to build the hit matrix H in each pixel, which is then inverted to compute the , defined in Eq. (A.6). Those are then multiplied together to build the scanning information matrix using its pixel space definition (Eq. (40b)). The resulting complex matrix contains 9n_{1}n_{2}(2s_{max} + 1) elements, where n_{1} and n_{2} are the number of detectors in each of the two detector assemblies whose crossspectra are considered. This step can be parallelized to a large extent, and can be dramatically sped up by building this matrix out of a representative subset of pixels. In our comparison to simulations, described in Sect. 5, and performed on HEALPix map with n_{side} = 2048 and pixels, we checked that using only N_{pix}/ 64 pixels evenly spread on the sky gave final results almost identical to those of the full calculations.
Finally, using Eqs. ((E.1)–(41)) we note that , so that, for instance, for a given ℓ, the 3x3 matrix is computed by replacing in Eq. (E.1) its central term C_{ℓ} with its partial derivative, such as where we assumed in Eq. (49) that, on the sky, and generally , like for CMB anisotropies. On the other hand, when dealing with arbitrary foregrounds crossfrequency spectra, we would have to assume when X′ ≠ Y′, and compute and separately. As we shall see in Sect. 6, this final and fastest step is the only one that needs to be repeated in a MonteCarlo analysis of instrumental errors, and it can be sped up. Indeed, since the input b_{ℓm} and output W_{ℓ} are generally very smooth functions of ℓ, it is not necessary to do this calculation for every single ℓ, but rather for a sparse subset of them, for instance regularly interspaced by δℓ. The resulting W_{ℓ} matrix is then Bspline interpolated. In our test cases with θ_{FWHM} = 10 to 5′, using δℓ = 10 leads to relative errors on the final product below 10^{5} for each ℓ.
In our tests, with s_{max} = 6, ℓ_{max} = 4000, n_{1} = n_{2} = 4, and all proposed speedups in place, Step 2 took about ten minutes, dominated by IO, while Step 3 took less than a minute on one core of a 3 GHz Intel Xeon CPU. The final product is a set of six (or nine) real matrices , each with 9(ℓ_{max} + 1) elements.
5. Comparison to PlanckHFI simulations
The differential nature of the polarization measurements, in the absence of modulating devices such as rotating halfwave plates, means that any mismatch between the responses of the two (or more) detectors being used will leak a fraction of temperature into polarization. This was observed in Planck, even though pairs of polarized orthogonal detectors observed the sky through the same horn, therefore with almost identical optical beams. Optical mismatches within pairs of detectors were enhanced by residuals of the electronic time response deconvolution which could affect their respective scanning beams differently (Planck Collaboration IV 2014; Planck Collaboration VII 2014). Other sources of mismatch included their different noise levels and thus their respective statistical weight on the maps, which could reach relative differences of up to 80%, and the number of valid samples which could vary by up to 20% between detectors. As seen previously, these detectorspecific features can be included in the QuickPol pipeline in order to describe as closely as possible the actual instrument. In this section, we show how we actually did it and how QuickPol compares to fullfledged simulations of PlanckHFI observations.
Noiseless simulations of PlanckHFI observations of a pure CMB sky were run for quadruplets of polarized detectors at three different frequencies (100, 143, and 217 GHz), and identified as 100ds1, 143ds1, and 217ds1 respectively. The input CMB power spectrum was assumed to contain no primordial tensorial modes, with the traditional and . The same mission duration, pointing, polarization orientations (γ_{j}) and efficiencies (ρ_{j}), flagged samples, and discarded pointing periods (f_{j}) were used as in the actual observations, with computer simulated polarized optical beams for the relevant detectors produced with the GRASP^{6} physical optics code (Rosset et al. 2007, and references therein) as illustrated on Fig. 4. Data streams were generated with the LevelS simulation pipeline (Reinecke et al. 2006), using the Conviqt code (Prézeau & Reinecke 2010) to perform the convolution of the sky with the beams, including the b_{ℓs} for  s  ≤ s_{max} = 14 and ℓ ≤ ℓ_{max} = 4800. Polarized maps of each detector set were produced with the Polkapix destriping code (Tristram et al. 2011), assuming the same noisebased relative weights (w_{j}) as the actual data, and their cross spectra were computed over the whole sky with HEALPix anafast routine to produce the empirical power spectra .
Fig. 4
Computer simulated beam maps (, , and clockwise from topleft) for two of the PlanckHFI detectors (1001a and 2175a) used in the validation of QuickPol. Each panel is 1°× 1° in size, and the units are arbitrary. 
Fig. 5
Comparison to simulations for 100ds1x217ds1 (lhs panels) and 143ds1x217ds1 (rhs panels) cross power spectra, for computer simulated beams. In each panel is shown the discrepancy between the actual ℓ(ℓ + 1)C_{ℓ}/ 2π and the one in input, smoothed on Δℓ = 31. Results obtained on simulations with either the full beam model (green curves) or the copolarized beam model (blue dashes) are to be compared to QuickPol analytical results (red long dashes). In panels where it does not vanish, a small fraction of the input power spectrum is also shown as black dots for comparison. 
The same exercise was reproduced replacing the initial beam maps with a purely copolarized beam based on the same , in order to test the validity of the copolarized assumption in Planck.
Figure 5 shows how the empirical power spectra are different from the input ones, (50)after correction from the pixel and (scalar) beam window functions, and compares those to the QuickPol predictions (51)for all nine possible values of XY for the crossspectra of detector sets 100ds1x217ds1 and 143ds1x217ds1. The results are actually multiplied by the usual ℓ(ℓ + 1)/2π factor, and smoothed on Δℓ = 31. The empirical results are shown both for the fullfledged beam model (green curves) and the purely copolarized model (blue dashes). One sees that the change, mostly visible in the EE case, is very small, validating the copolarized beam assumption, at least within the limits of this computer simulated Planck optics. The QuickPol predictions, only shown in the copolarized case for clarity (long red dashes), agree extremely well with the corresponding numerical simulations. We have checked that this agreement to simulations remains true in the full beam model.
6. Propagation of instrumental uncertainties
We assumed so far the instrument to be nonideal, but exactly known. In practice, however, the instrument is only known with limited accuracy and the final beam matrix will be affected by at least four types of uncertainties:

limited knowledge of the beam angular response, which affects the , replacing them with while preserving the beam total throughput after calibration (see below) . We therefore assume the beam power spectrum W_{ℓ} = ∑ _{m}  b_{ℓm}  ^{2}/ (2ℓ + 1) to be the same at ℓ = 0, where the beam throughput is defined, and at ℓ = 1, where the detector gain calibration is usually done using the CMB dipole.

error on the gain calibration of detector j, which translates into , with  δc_{j}  ≪ 1,

error on the polar efficiency of detector j, which translates into . As discussed in Sect. 2.1, we expect in the case of PlanckHFI a relative uncertainty  δρ_{j}/ρ_{j}  < 1%.

error on the actual direction of polarization: for each detector j, the direction of polarization measured in a common referential becomes γ_{j}− → γ_{j} + δγ_{j}. In the case of PlanckHFI, Rosset et al. (2010) found the preflight measurement of this angle to be dominated by systematic errors of the order of 1° for polarization sensitive bolometers (PSBs). These uncertainties can be larger for spider web bolometers (SWBs), but as we shall see below, the coupling with the low polarization efficiency ρ_{j} of those detectors makes them somewhat irrelevant.
All these uncertainties can be inserted in Eq. (E.1) by substituting Eq. (E.4) with (52)where x_{j} = (1 + δρ_{j}/ρ_{j})e^{2iδγj}.
As mentioned in Sect. 3, such substitutions are done in Step 3 of the QuickPol pipeline. A new set of numerical values for the instrument model can therefore be turned rapidly into a beam window matrix (Eq. (41)), allowing, for instance, a MonteCarlo exploration at the power spectrum level of the instrumental uncertainties.
7. About rotating halfwave plates
In the previous sections we have focused on experiments that rely on the rotation of the full instrument with respect to the sky to have the angular redundancy required to measure the Stokes parameters. An alternative way is to rotate the incoming polarization at the entrance of the instrument while leaving the rest fixed. This is most conveniently achieved with a rotating halfwave plate (rHWP). The rotation is either stepped (POLARBEAR Collaboration 2014) or continuous (Chapman et al. 2014; EssingerHileman et al. 2016; Ritacco et al. 2016a). The advantages of this system are numerous, the first of which is the decoupling between the optimization of the scanning strategy in terms of “pure” redundancy and its optimization in terms of “angular” redundancy. It is much easier to control the rotation of a rHWP than of a full instrument and therefore ensure an optimal angular coverage whatever the observation scene is. If the rotation is continuous and fast, typically of the order of 1 Hz, it has the extra advantage of modulating polarization at frequencies larger than the atmospheric and electronics 1 /f noise knee frequency, hence ensuring a natural rejection of these low frequency noises. Furthermore, this allows us to build I, Q, and U maps per detector, without needing to combine different detectors with their associated bandpass mismatch or other differential systematic effects mentioned in the previous sections. Individual detector systematics therefore tend to average out rather than combine to induce leakage between sky components. On the down side, this comes at the price of moving a piece of hardware in the instrument and all its associated systematic effects, starting with a signal that is synchronous with the rHWP rotation as observed in Johnson et al. (2007), Chapman et al. (2014), and Ritacco et al. (2016b).
Such tradeoffs are being investigated by current experiments using rHWPs and will certainly be studied in more details in preparation of future CMB orbital and suborbital missions, such as CMBS4 network (CMBS4 Collaboration 2016). We here briefly comment on how the addition of a rHWP to an instrument can be coped with in QuickPol.
The Jones matrix of a HWP (which shifts the yaxis electric field by a half period) rotated by an angle ψ is (O’Dea et al. 2007) If a rotating HWP is installed at the entrance of the optical system, the Jones matrix of the system becomes , and the signal observed in the presence of arbitrary beams (Eq. (8)) becomes (after dropping the circular polarization V terms) (55)with These new beams can then be passed to Eq. (30) and propagated through the rest of QuickPol. Together with Eq. (9), we see that, if ψ is correctly chosen, the modulation of Q and U, by 2α + 4ψ, is now clearly different from that of T which depends only on α via r_{α}, even for noncircular beams. The leakages from temperature to polarization are therefore expected to be much smaller than when the polarization modulation is performed only by a rotation of the whole instrument, and O’Dea et al. (2007) showed, that even for nonideal rHWP, the induced systematic effects are limited to polarization crosstalks without temperature to polarization leakage.
As previously mentioned, specific systematic effects such as the rotation synchronous signal must be treated with care. Once such time domain systematic effects are identified and modeled, they, together with realistic optical properties of the instrument, can be integrated in the QuickPol formalism in order to be taken into account, quantified, and/or marginalized over at the power spectrum level.
8. Conclusions
Polarization measurements are mostly obtained by differencing observations by different detectors. Mismatch in their optical beams, time responses, bandpasses, and so on induces systematic effects, for example, temperature to polarization leakage. The QuickPol formalism allows us to compute accurately and efficiently the induced crosstalk between temperature and polarization power spectra. It also provides a fast and easy way to propagate instrumental modeling uncertainties down to the final angular power spectra and is thus a powerful tool to simulate observations and to help with the design and specifications of future experiments, such as acceptable beam distortions, polarization modulation optimization, and observation redundancy. It can cope with time varying instrumental parameters, realistic sample flagging, and rejection. The method was validated through comparison to numerical simulations of realistic Planck observations. The hypotheses required on the instrument and survey, described in Sects. 2 and 3, are extremely general and apply to Planck and to forthcoming CMB experiments such as PIXIE, LiteBIRD, COrE, and others. Contrary to MonteCarlo based methods, such as FEBeCoP, the impact of the beam related imperfections on the measured power spectra are obtained without having to assume any prior knowledge of the sky power spectra.
Of course, the beam matrices provided by QuickPol can be used in the cosmological analysis of a CMB survey. Indeed, the sky power spectra can be modeled as functions of cosmological parameters { θ_{C} }, foreground modeling { θ_{F} }, and nuisance parameters { θ_{n} }. These can then be generated, multiplied with the beam matrices for the set of detectors being analyzed, and compared to the measured in a maximum likelihood sense, in the presence of instrumental noise. The parameters { θ_{C} } , { θ_{F} } , { θ_{n} } can be iterated or integrated upon, with statistical priors, until a posterior distribution is built. In this kind of forward approach, it is not necessary to correct the observations from possibly singular transfer functions, nor to backpropagate the noise. At least some of the instrumental uncertainties { θ_{I} } affecting the effective beam via could be included in the overall analysis, and marginalized over, thanks to the fast calculation times by QuickPol of the impact of changes in the gain calibrations, polarization angles, and efficiencies, as discussed in Sect. 6. While QuickPol has been originally developed and tested in the case of experiments without a rotating halfwave plate, it is straightforward to add one to the current pipeline and assess its impact on the aforementioned systematics. Specific additional effects such as a HWP rotation synchronous signal or the effect of a tilted HWP are expected to show up in real experiments. As long as these can be physically modeled, they can be inserted in QuickPol as well.
Wilkinson microwave anisotropy probe: http://map.gfsc.nasa.gov.
Although it is important when trying to disentangle sky signals with different electromagnetic spectra (Planck Collaboration VI 2014), the finite bandwidth of the actual detectors only plays a minor role in the problem considered here, and will be ignored in this paper.
TICRA: http://www.ticra.com.
Acknowledgments
Thank you to the Planck collaboration, and in particular to D. Hanson, K. Benabed, and F. R. Bouchet for fruitful discussions. Some results presented here were obtained with the HEALPix library.
References
 ArmitageCaplan, C., & Wandelt, B. D. 2009, ApJS, 181, 533 [NASA ADS] [CrossRef] [Google Scholar]
 Bennett, C. L., Larson, D., Weiland, J. L., et al. 2013, ApJS, 208, 20 [NASA ADS] [CrossRef] [Google Scholar]
 Bunn, E. F., Zaldarriaga, M., Tegmark, M., & de OliveiraCosta, A. 2003, Phys. Rev. D, 67, 023501 [NASA ADS] [CrossRef] [Google Scholar]
 Challinor, A., Fosalba, P., Mortlock, D., et al. 2000, Phys. Rev. D, 62, 123002 [NASA ADS] [CrossRef] [Google Scholar]
 Chapman, D., Aboobaker, A. M., Ade, P., et al. 2014, Am. Astron. Soc. Meet. Abstr., 223, 407.03 [Google Scholar]
 Chon, G., Challinor, A., Prunet, S., Hivon, E., & Szapudi, I. 2004,MNRAS, 350, 914 [Google Scholar]
 CMBS4 Collaboration 2016, ArXiv eprints [arXiv:1610.02743] [Google Scholar]
 Edmonds, A. R. 1957, Angular Momentum in Quantum Mechanics (Princeton University Press) [Google Scholar]
 EssingerHileman, T., Kusaka, A., Appel, J. W., et al. 2016, Rev. Sci. Inst., 87, 094503 [NASA ADS] [CrossRef] [Google Scholar]
 Fosalba, P., Doré, O., & Bouchet, F. R. 2002, Phys. Rev. D, 65, 063003 [NASA ADS] [CrossRef] [Google Scholar]
 Górski, K. M., Hivon, E., Banday, A. J., et al. 2005, ApJ, 622, 759 [NASA ADS] [CrossRef] [Google Scholar]
 Grain, J., Tristram, M., & Stompor, R. 2009, Phys. Rev. D, 79, 123515 [NASA ADS] [CrossRef] [Google Scholar]
 Hansen, F. K., & Górski, K. M. 2003, MNRAS, 343, 559 [NASA ADS] [CrossRef] [Google Scholar]
 Hanson, D., Lewis, A., & Challinor, A. 2010, Phys. Rev. D, 81, 103003 [NASA ADS] [CrossRef] [Google Scholar]
 Hinshaw, G., Nolta, M. R., Bennett, C. L., et al. 2007, ApJS, 170, 288 [NASA ADS] [CrossRef] [Google Scholar]
 Hivon, E., Górski, K. M., Netterfield, C. B., et al. 2002, ApJ, 567, 2 [NASA ADS] [CrossRef] [Google Scholar]
 Hu, W. 2000, Phys. Rev. D, 62, 043007 [NASA ADS] [CrossRef] [Google Scholar]
 Hu, W., Hedman, M. M., & Zaldarriaga, M. 2003, Phys. Rev. D, 67, 043004 [NASA ADS] [CrossRef] [Google Scholar]
 Johnson, B. R., Collins, J., Abroe, M. E., et al. 2007, ApJ, 665, 42 [NASA ADS] [CrossRef] [Google Scholar]
 Keihänen, E., & Reinecke, M. 2012, A&A, 548, A110 [NASA ADS] [CrossRef] [EDP Sciences] [Google Scholar]
 Keihänen, E., Kiiveri, K., KurkiSuonio, H., & Reinecke, M. 2016, ArXiv eprints [arXiv:1610.00962] [Google Scholar]
 Leahy, J. P., Bersanelli, M., D’Arcangelo, O., et al. 2010, A&A, 520, A8 [NASA ADS] [CrossRef] [EDP Sciences] [Google Scholar]
 Lewis, A., & Challinor, A. 2006, Phys. Rep., 429, 1 [NASA ADS] [CrossRef] [EDP Sciences] [Google Scholar]
 Miller, N. J., Shimon, M., & Keating, B. G. 2009a, Phys. Rev. D, 79, 063008 [NASA ADS] [CrossRef] [Google Scholar]
 Miller, N. J., Shimon, M., & Keating, B. G. 2009b, Phys. Rev. D, 79, 103002 [NASA ADS] [CrossRef] [Google Scholar]
 Mitra, S., Sengupta, A. S., & Souradeep, T. 2004, Phys. Rev. D, 70, 103002 [NASA ADS] [CrossRef] [Google Scholar]
 Mitra, S., Sengupta, A. S., Ray, S., Saha, R., & Souradeep, T. 2009, MNRAS, 394, 1419 [NASA ADS] [CrossRef] [Google Scholar]
 Mitra, S., Rocha, G., Górski, K. M., et al. 2011, ApJS, 193, 5 [NASA ADS] [CrossRef] [Google Scholar]
 O’Dea, D., Challinor, A., & Johnson, B. R. 2007, MNRAS, 376, 1767 [NASA ADS] [CrossRef] [Google Scholar]
 Page, L., Hinshaw, G., Komatsu, E., et al. 2007, ApJS, 170, 335 [NASA ADS] [CrossRef] [Google Scholar]
 Pant, N., Das, S., Rotti, A., Mitra, S., & Souradeep, T. 2016, J. Cosmol. Astropart. Phys., 3, 035 [NASA ADS] [CrossRef] [Google Scholar]
 Planck Collaboration IV. 2014, A&A, 571, A4 [NASA ADS] [CrossRef] [EDP Sciences] [Google Scholar]
 Planck Collaboration VI. 2014, A&A, 571, A6 [NASA ADS] [CrossRef] [EDP Sciences] [Google Scholar]
 Planck Collaboration VII. 2014, A&A, 571, A7 [NASA ADS] [CrossRef] [EDP Sciences] [Google Scholar]
 Planck Collaboration VIII. 2014, A&A, 571, A8 [NASA ADS] [CrossRef] [EDP Sciences] [Google Scholar]
 Planck Collaboration XVII. 2014, A&A, 571, A17 [NASA ADS] [CrossRef] [EDP Sciences] [Google Scholar]
 Planck Collaboration VII. 2016, A&A, 594, A7 [NASA ADS] [CrossRef] [EDP Sciences] [Google Scholar]
 Planck Collaboration XI. 2016, A&A, 594, A11 [NASA ADS] [CrossRef] [EDP Sciences] [Google Scholar]
 Polarbear Collaboration 2014, ApJ, 794, 171 [NASA ADS] [CrossRef] [Google Scholar]
 Prézeau, G., & Reinecke, M. 2010, ApJS, 190, 267 [NASA ADS] [CrossRef] [Google Scholar]
 Ramamonjisoa, F. A., Ray, S., Mitra, S., & Souradeep, T. 2013, ArXiv eprints [arXiv:1309.4784] [Google Scholar]
 Rathaus, B., & Kovetz, E. D. 2014, MNRAS, 443, 750 [NASA ADS] [CrossRef] [Google Scholar]
 Reinecke, M., Dolag, K., Hell, R., Bartelmann, M., & Enßlin, T. A. 2006, A&A, 445, 373 [NASA ADS] [CrossRef] [EDP Sciences] [Google Scholar]
 Ritacco, A., Adam, R., Adane, A., et al. 2016a, ArXiv eprints [arXiv:1602.01605] [Google Scholar]
 Ritacco, A., Ponthieu, N., Catalano, A., et al. 2016b, ArXiv eprints [arXiv:1609.02042] [Google Scholar]
 Rosset, C., Yurchenko, V., Delabrouille, J., et al. 2007, A&A, 464, 405 [NASA ADS] [CrossRef] [EDP Sciences] [Google Scholar]
 Rosset, C., Tristram, M., Ponthieu, N., et al. 2010, A&A, 520, A13 [NASA ADS] [CrossRef] [EDP Sciences] [Google Scholar]
 Shimon, M., Keating, B., Ponthieu, N., & Hivon, E. 2008, Phys. Rev. D, 77, 83003 [NASA ADS] [CrossRef] [Google Scholar]
 Smith, K. M., Zahn, O., & Doré, O. 2007, Phys. Rev. D, 76, 043510 [NASA ADS] [CrossRef] [Google Scholar]
 Souradeep, T., & Ratra, B. 2001, ApJ, 560, 28 [NASA ADS] [CrossRef] [Google Scholar]
 Tristram, M., MacíasPérez, J. F., Renault, C., & Hamilton, J.C. 2004, Phys. Rev. D, 69, 123008 [NASA ADS] [CrossRef] [Google Scholar]
 Tristram, M., Filliard, C., Perdereau, O., et al. 2011, A&A, 534, A88 [NASA ADS] [CrossRef] [EDP Sciences] [Google Scholar]
 Wallis, C. G. R., Brown, M. L., Battye, R. A., Pisano, G., & Lamagna, L. 2014, MNRAS, 442, 1963 [NASA ADS] [CrossRef] [Google Scholar]
 Wallis, C. G. R., Bonaldi, A., Brown, M. L., & Battye, R. A. 2015, MNRAS, 453, 2058 [NASA ADS] [CrossRef] [Google Scholar]
 Zaldarriaga, M., & Seljak, U. 1997, Phys. Rev. D, 55, 1830 [NASA ADS] [CrossRef] [Google Scholar]
Appendix A: Projection of maps on spherical harmonics
Here we give more details on the steps required to go from Eq. (35) to Eq. (38). Let us recall Eq. (35) and explain it further:
where we introduced the sth complex moment of the direction of polarization for detector j, (A.3)the hit matrix H defined for (u,v) ∈ { 0,2,−2 } ^{2} as (A.4)with (A.5)the hit normalized moments (A.6)which are described in Appendix B, and finally the inverse noise variance weighted beam spherical harmonics (SH) coefficients Since the solution of Eq. (33) remains the same when all the noise covariances are rescaled simultaneously by an arbitrary factor a: N_{j}− → aN_{j}, one can also rescale the weights w_{j} appearing in Eqs (A.4) and (A.7), with for instance w_{j}− → w_{j}/ ∑ _{k}w_{k} without altering the final result.
The components of the observed polarized map are then (A.9)with (A.10)After expansion of the hit normalized moments (Eq. (A.6)) in spherical harmonics: (A.11)the polarized map reads (A.12)and the SH coefficients of spin x of map are, for pixels of area Ω_{p}, which are only nonzero when x = v. The cross power spectrum of spin v_{1} and v_{2} maps is then given by Eq. (38).
Appendix B: Hit matrix
Introducing, for detector j, (B.1)the Hermitian hit matrix for a weighted combination of detectors is with h,x real and z_{2},z_{4} complex numbers, and has for inverse (B.4)with (B.5)In Eq. (A.6) we defined (B.6)for any value of s, which provides (B.7)so that is of spin s, provided z_{2} and z_{4} are of spin 2 and 4 respectively. Since , we get .
By definition, (B.8)so that (B.9)
Appendix C: Wigner 3J symbols
The Wigner 3J symbols describe the coupling between different spin weighted spherical harmonics at the same location: (C.1)and the symbol is nonzero only when,  m_{i}  ≤ ℓ_{i} for i = 1,2,3, m_{1} + m_{2} + m_{3} = 0 and (C.2)They obey the relations (C.3)and (C.4)Their standard orthogonality relations are (C.5)and (C.6)where δ(ℓ_{1},ℓ_{2},ℓ_{3}) = 1 when ℓ_{1},ℓ_{2},ℓ_{3} obey the triangle relation of Eq. (C.2) and vanishes otherwise.
For ℓ_{1} ≪ ℓ_{2},ℓ_{3} (Edmonds 1957, Eq. (A2.1)) (C.7)where d is the Wigner rotation matrix and cosθ = 2m_{2}/ (2ℓ_{2} + 1). As a consequence, for  m_{2}  ≪ ℓ_{2}(C.8)and an approximate orthogonality relation can therefore be written, for ℓ_{1},  m_{1}  ,  m_{2}  ≪ ℓ_{2},ℓ_{3}(C.9)
Appendix D: Spin weighted power spectra
Since a complex field of spin s can be written as C_{s} = R_{s} + iI_{s} where R_{s} and I_{s} are real, with (D.1)and, with the CondonShortley phase convention then (D.2)When s = 2, one defines such that , with X = E,B, and When s = 1, one defines such that , with X = G,C, and
Appendix E: Window matrices W
Appendix E.1: Arbitrary beams, smooth scanning case
Let us come back to Eqs. (39) and (40). These can be cast in a more compact matrix form (E.1)where (E.2)(E.3)(E.4)and X ∗ Y denotes the elementwise product (also known as Hadamard or Schur product) of arrays X and Y. Noting that (E.5)where R_{2} was introduced in Eq. (18), which leads to Eq. (41) that we recall here for convenience: (E.6)Introducing the shorthand (E.7)describing the coupled moments of the polarized detectors j_{1} and j_{2} orientation, and assuming in Eq. (E.4) the beams to be perfectly copolarized, with polar efficiencies , one gets, for XY = TT,EE,BB,TE,TB,EB,ET,BT,BE: (E.8a)which is illustrated in Fig. 3; (E.8b)(E.8c)Since, by definition (Eqs. (A.7) and (E.7)), and , one can check that each term of is real, as expected.
Appendix E.2: Arbitrary beams, ideal scanning
In the case of ideal scanning described in Sect. 3.4, one gets , so that the hit matrix is diagonal: (E.9)and the orientation moments (E.10)are such that (E.11)with (E.12)One then obtains the beam matrices (E.13a)(E.13b)(E.13c)The implications of Eq. (E.13) are discussed in Sect. 3.4.
Appendix F: Finite pixel size
Introducing the spin raising and lowering differential operators applied to a function f of spin s, (Zaldarriaga & Seljak 1997; Bunn et al. 2003, and references therein) the spin weighed spherical harmonics are defined as such that with .
As noticed in Planck Collaboration VII (2014) and Planck Collaboration XVII (2014), the formalism of subpixel effect is very close to the one of gravitational lensing described in Hu (2000) and Lewis & Challinor (2006).
For r = (1,θ,ϕ) = e_{r} and dr = (0,dθ,dϕ) = dθe_{θ} + sinθdϕe_{ϕ}, with dr = dr.(e_{θ} + ie_{ϕ}) = dθ + isinθdϕ, and g(ℓ,s) = f(ℓ,s)f(ℓ,s + 1).
Identifying dr to the position of a measurement relative to the nominal center r of the pixel to which it is attributed, this expansion of can be injected into Eqs. (28) and (A.14). Assuming dr to be uncorrelated with the orientation of the detector, two extra terms, both quadratic in dr, will appear in the final power spectra.
The first term involves the scalar product of the gradient of the signal in the pixel, assumed to be totally dominated by the temperature, with the weighted sum of dr over all samples in that pixel. Introducing (F.10)which is of spin s + v ± 1 and such that , one finds (F.11)with In the case of temperature, and assuming the beams to be circular, this simplifies to (F.14)in agreement with Planck Collaboration VII (2014), once one identifies as the sum of the gradient and curl parts of the (spin 1) displacement field power spectrum and as their difference (see Sect. D).
It is instructive to further assume the relative location of the hit’s center of mass to be only weakly correlated between pixels, so that all its derived power spectra can be assumed to be white: (i.e., with a variance in pixels of solid angle Ω_{pix} = 4π/N_{pix}). Equation (C.5) then ensures that the subpixel noise of Eq. (F.11) is also white, with constant polarized spectra (F.15)with where the approximate results are obtained for circular beams.
Even for more realistic hypotheses on the hits locations, the subpixel contributions to the respective power spectra follow the hierarchy (F.21)Let us consider now the other extra contribution to the power spectrum, involving the Laplacian of the sky signal and the quadratic norm of dr. Introducing where is the second order moment of the hit location in pixel p. If we assume this and to be slowly varying functions of p, and consider ℓ ≫ s, the power spectra become (F.24)which describes to leading order, the smoothing effect of the integration of the signal on the pixel.
Appendix G: Copolarized beam
For an arbitrarily shaped beam having the intensity harmonics coefficients (G.1)and assumed to be perfectly copolarized in direction γ, its polarized harmonics’ content will be where we used Eq. (C.1) and introduced, for ℓ′′ ≥ 2Since I_{ℓ′′} peaks at ℓ′′ = 2, the 3J symbols will enforce ℓ′′ ≪ ℓ ≃ ℓ′ in Eq. (G.4). If the beam is narrow enough in real space, b_{ℓ′,m} will be almost constant over the allowed range ℓ−ℓ′′ ≤ ℓ′ ≤ ℓ + ℓ′′, and we use the approximate orthogonality relation of Eq. (C.9) to write (G.8)Finally, we note that in Eq. (G.4), the sum to obtain (G.12)which is valid for any (narrow) copolarized beam.
All Figures
Fig. 1
Orientation of polarization measurements in Planck. The two left panels show, for an actual Planck detector, the maps of ⟨cos2α⟩ and ⟨sin2α⟩ respectively, where α is the direction of the polarizer with respect to the local Galactic meridian, which contributes to the spin 2 term defined in Eq. (A.3). The right panel shows the power spectrum of , multiplied by ℓ(ℓ + 1)/2π. 

In the text 
Fig. 2
Same as Fig. 1 for an hypothetical detector of a LiteBIRDlike mission, except for the right panel plot which has a different yrange. 

In the text 
Fig. 3
Effective beam window matrix introduced in Eq. (41) and detailed in Eq. (E.8a), for the crossspectra of two simulated Planck maps discussed in Sect. 5. Left panel: raw elements of , showing for each ℓ how the measured XY map angular power spectrum is impacted by the input TT spectrum, because of the observation of the sky with the beams. Right panel: blownup ratio of the nondiagonal elements to the diagonal ones: 100 W^{XY,TT}_{ℓ}/W^{TT,TT}_{ℓ} 

In the text 
Fig. 4
Computer simulated beam maps (, , and clockwise from topleft) for two of the PlanckHFI detectors (1001a and 2175a) used in the validation of QuickPol. Each panel is 1°× 1° in size, and the units are arbitrary. 

In the text 
Fig. 5
Comparison to simulations for 100ds1x217ds1 (lhs panels) and 143ds1x217ds1 (rhs panels) cross power spectra, for computer simulated beams. In each panel is shown the discrepancy between the actual ℓ(ℓ + 1)C_{ℓ}/ 2π and the one in input, smoothed on Δℓ = 31. Results obtained on simulations with either the full beam model (green curves) or the copolarized beam model (blue dashes) are to be compared to QuickPol analytical results (red long dashes). In panels where it does not vanish, a small fraction of the input power spectrum is also shown as black dots for comparison. 

In the text 
Current usage metrics show cumulative count of Article Views (fulltext article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 4896 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.