This article has an erratum: [erratum]
Volume 612, April 2018
|Number of page(s)||12|
|Published online||13 April 2018|
SEPIA – a new single pixel receiver at the APEX telescope
Group for Advanced Receiver Development (GARD), Department of Space, Earth and Environment, Chalmers University of Technology,
2 Department of Space, Earth and Environment, Chalmers University of Technology, Onsala Space Observatory (OSO), 43992 Onsala, Sweden
3 European Southern Observatory (ESO), Karl-Schwarzschild-Str. 2, 85748 Garching bei München, Germany
4 European Southern Observatory, Alonso de Córdova 3107, Vitacura, Casilla 19001, Santiago de Chile, Chile
5 Netherlands Research School for Astronomy (NOVA), Kapteyn Astronomical Institute, Landleven 12, 9747 AD Groningen, The Netherlands
Accepted: 19 November 2017
Context. We describe the new Swedish-ESO PI Instrument for APEX (SEPIA) receiver, which was designed and built by the Group for Advanced Receiver Development (GARD), at Onsala Space Observatory (OSO) in collaboration with ESO. It was installed and commissioned at the APEX telescope during 2015 with an ALMA Band 5 receiver channel and updated with a new frequency channel (ALMA Band 9) in February 2016.
Aim. This manuscript aims to provide, for observers who use the SEPIA receiver, a reference in terms of the hardware description, optics and performance as well as the commissioning results.
Methods. Out of three available receiver cartridge positions in SEPIA, the two current frequency channels, corresponding to ALMA Band 5, the RF band 158–211 GHz, and Band 9, the RF band 600–722 GHz, provide state-of-the-art dual polarization receivers. The Band 5 frequency channel uses 2SB SIS mixers with an average SSB noise temperature around 45 K with IF (intermediate frequency) band 4–8 GHz for each sideband providing total 4 × 4 GHz IF band. The Band 9 frequency channel uses DSB SIS mixers with a noise temperature of 75–125 K with IF band 4–12 GHz for each polarization.
Results. Both current SEPIA receiver channels are available to all APEX observers.
Key words: instrumentation: detectors / techniques: spectroscopic
© ESO 2018
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.