Volume 490, Number 3, November II 2008
|Page(s)||1157 - 1163|
|Published online||17 September 2008|
A Swedish heterodyne facility instrument for the APEX telescope
Onsala Space Observatory (OSO), Chalmers University of Technology, 43992, Sweden e-mail: email@example.com
2 European Southern Observatory (ESO), Casilla 19001, Santiago 19, Chile
Accepted: 25 August 2008
Aims. In March 2008, the APEX facility instrument was installed on the telescope at the site of Lliano Chajnantor in northern Chile. The main objective of the paper is to introduce the new instrument to the radio astronomical community. It describes the hardware configuration and presents some initial results from the on-sky commissioning.
Methods. The heterodyne instrument covers frequencies between 211 GHz and 1390 GHz divided into four bands. The first three bands are sideband-separating mixers operating in a single sideband mode and based on superconductor-insulator-superconductor (SIS) tunnel junctions. The fourth band is a hot-electron bolometer, waveguide balanced mixer. All bands are integrated in a closed-cycle temperature-stabilized cryostat and are cooled to 4 K.
Results. We present results from noise temperature, sideband separation ratios, beam, and stability measurements performed on the telescope as a part of the receiver technical commissioning. Examples of broad extragalactic lines are also included.
Key words: telescopes / instrumentation: detectors / techniques: spectroscopic / radio lines: general
© ESO, 2008
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.