Drift of HARPS as a function of time measured on the Th-Ar spectra taken each night to derive the wavelength solution of the detector. Each subplot represents the drift in RVs measured on the Th-Ar emission lines falling in different regions of the HARPS detector. The HARPS detector is cut into 24 regions, 16 corresponding to the blue CCD shown with a blue background (pixel 0–1024, 1024–2048, 2048–3072, 3072–4096 and orders 0–12, 12–24, 24–36 and 36–47) and 8 corresponding to the red CCD shown with a red background (same separation in pixel but for order 47–60 and 60–72). The main feature that we see here is the huge RV offset on JD = 2457174 (1st of June 2015), which is induced by the change of the HARPS fibres from circular to octagonal.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.