Volume 515, June 2010
|Number of page(s)||10|
|Published online||10 June 2010|
First step to detect an extrasolar planet using simultaneous observations with the VLTI instruments AMBER and MIDI*
Laboratoire Fizeau, UMR 6525, UNS - Observatoire de la Côte d'Azur, BP 4229, 06304 Nice Cedex 4, France e-mail: email@example.com
2 Laboratoire Fizeau, UMR 6525, UNS - Observatoire de la Côte d'Azur, 06108 Nice Cedex 02, France
3 ESO, Casilla 19001, Vitacura, Santiago 19, Chile
4 Leiden Observatory, PO Box 9513, 2300 RA, Leiden, The Netherlands
5 MPIA, Königstuhl 17, 69117 Heidelberg, Germany
Accepted: 8 January 2010
Aims. Performed in November 2007 as a part of the MIDI Guaranteed Time Observation Exoplanet Programme, the observation of Gliese 86b constituted the first attempt at an exoplanet detection with the VLTI instrument MIDI. It is also a technical achievement since it motivated the first VLTI observation using AMBER and MIDI simultaneously.
Methods. Fringes were obtained for both instruments with the aim of reaching sufficient precision on the low differential phase signal of Gliese 86b. The principle is to correct the phase measured in N-band from the water vapour dispersion using the fringes in K-band. In N-band, the source, Gliese 86, has an estimated magnitude of 3.8. With a separation of 0.11 AU, a flux ratio of about 10-3 is expected between the planet and the star. According to the measurement principle and the planet signal signature, the effective expected interferometric phase is a curved-like function of the wavelength with a mean amplitude of about 0.03°.
Results. Based on the MIDI phase measurements of the calibrator HD 9362, our study shows that a precision on the curvature measurement of about 0.33° is currently reached. Consequently, we stand at a factor 10 above the phase signal from the planet. The AMBER data, obtained in parallel, were too noisy to extrapolate and to remove the corresponding dispersion in N band at the required level of precision. However, we report the set of data obtained, we discuss the calibration process involved, and we estimate its theoretical efficiency.
Key words: instrumentation: interferometers / techniques: interferometric / planetary systems / infrared: general / atmospheric effects
© ESO, 2010
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.