Schematics of the MpI concept. Two options are shown for the reference source: (a) a de-focused image of the instantaneous field-of-view; and (b) a cold reference. The polariser P1 provides the needed polarised input to the MpI. P2 is the beam splitter defining the two arms, while P3, in the cryostat, dispatches the two projections of the polarised signal to the focal-plane arrays. The incoming beam is represented by a spectral distribution S(ν).
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.