Volume 600, April 2017
|Number of page(s)||9|
|Published online||28 March 2017|
On-sky performance of the QACITS pointing control technique with the Keck/NIRC2 vortex coronagraph
1 Space Sciences, Technologies, and Astrophysics Research (STAR) Institute, Université de Liège, 19c Allée du Six Août, 4000 Liège, Belgium
2 Jet Propulsion Laboratory, California Institute of Technology, 4800 Oak Grove Drive, Pasadena, CA 91109, USA
3 W. M. Keck Observatory, 65-1120 Mamalahoa Hwy., Kamuela, HI 96743, USA
4 California Institute of Technology, Division of Geological and Planetary Sciences, 1200 E. California Blvd, Pasadena, CA 91125, USA
5 California Institute of Technology, Division of Physics, Mathematics and Astronomy, 1200 E. California Blvd, Pasadena, CA 91125, USA
Received: 12 December 2016
Accepted: 20 January 2017
Context. A vortex coronagraph is now available for high contrast observations with the Keck/NIRC2 instrument at L band. The vortex coronagraph uses a vortex phase mask in a focal plane and a Lyot stop in a downstream pupil plane to provide high contrast at small angular separations from the observed host star.
Aims. Reaching the optimal performance of the coronagraph requires fine control of the wavefront incident on the phase mask. In particular, centering errors can lead to significant stellar light leakage that degrades the contrast performance and prevents the observation of faint planetary companions around the observed stars. It is thus critical to correct for the possible slow drift of the star image from the phase mask center, generally due to mechanical flexures induced by temperature and/or gravity field variation, or to misalignment between the optics that rotate in pupil tracking mode.
Methods. A control loop based on the QACITS algorithm for the vortex coronagraph has been developed and deployed for the Keck/NIRC2 instrument. This algorithm executes the entire observing sequence, including the calibration steps, initial centering of the star on the vortex center, and stabilisation during the acquisition of science frames.
Results. On-sky data show that the QACITS control loop stabilizes the position of the star image down to 2.4 mas rms at a frequency of about 0.02 Hz. However, the accuracy of the estimator is probably limited by a systematic error due to a misalignment of the Lyot stop with respect to the entrance pupil, estimated to be on the order of 4.5 mas. A method to reduce the amplitude of this bias down to 1 mas is proposed.
Conclusions. The QACITS control loop has been successfully implemented and provides a robust method to center and stabilize the star image on the vortex mask. In addition, QACITS ensures a repeatable pointing quality and significantly improves the observing efficiency compared to manual operations. It is now routinely used for vortex coronagraph observations at Keck/NIRC2, providing contrast and angular resolution capabilities suited for exoplanet and disk imaging.
Key words: instrumentation: adaptive optics / techniques: high angular resolution / methods: observational
© ESO, 2017
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.