Volume 569, September 2014
|Number of page(s)||9|
|Published online||26 September 2014|
A laser-lock concept to reach cm s-1-precision in Doppler experiments with Fabry-Pérot wavelength calibrators
Institut für Astrophysik,
2 IV. Physikalisches Institut, Friedrich-Hund-Platz 1, 37077 Göttingen, Germany
Accepted: 6 August 2014
State-of-the-art Doppler experiments require wavelength calibration with precision at the cm s-1 level. A low-finesse Fabry-Pérot interferometer (FPI) can provide a wavelength comb with a very large bandwidth as required for astronomical experiments, but unavoidable spectral drifts are difficult to control. Instead of actively controlling the FPI cavity, we propose to passively stabilize the interferometer and track the time-dependent cavity length drift externally using the 87Rb D2 atomic line. A dual-finesse cavity allows drift tracking during observation. In the low-finesse spectral range, the cavity provides a comb transmission spectrum tailored to the astronomical spectrograph. The drift of the cavity length is monitored in the high-finesse range relative to an external standard: a single narrow transmission peak is locked to an external cavity diode laser and compared to an atomic frequency from a Doppler-free transition. Following standard locking schemes, tracking at sub-mm s-1 precision can be achieved. This is several orders of magnitude better than currently planned high-precision Doppler experiments, and it allows freedom for relaxed designs including the use of a single-finesse interferometer under certain conditions. All components for the proposed setup are readily available, rendering this approach particularly interesting for upcoming Doppler experiments. We also show that the large number of interference modes used in an astronomical FPI allows us to unambiguously identify the interference mode of each FPI transmission peak defining its absolute wavelength solution. The accuracy reached in each resonance with the laser concept is then defined by the cavity length that is determined from the one locked peak and by the group velocity dispersion. The latter can vary by several 100 m s-1 over the relevant frequency range and severely limits the accuracy of individual peak locations, although their interference modes are known. A potential way to determine the absolute peak positions is to externally measure the frequency of each individual peak with a laser frequency comb (LFC). Thus, the concept of laser-locked FPIs may be useful for applying the absolute accuracy of an LFC to astronomical spectrographs without the need for an LFC at the observatory.
Key words: techniques: radial velocities / instrumentation: spectrographs / planets and satellites: detection / techniques: spectroscopic
© ESO, 2014
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.