Volume 490, Number 2, November I 2008
|Page(s)||859 - 871|
|Published online||11 September 2008|
SYMPA, a dedicated instrument for Jovian seismology
II. Real performance and first results
Laboratoire FIZEAU, Université de Nice Sophia-Antipolis, CNRS-Observatoire de la Côte d'Azur, 06108 Nice Cedex 2, France e-mail: Patrick.Gaulme@obspm.fr
2 Observatorio Astronómico Nacional, Instituto de Astronomía, Universidad Nacional Autónoma de México, Apto. Postal 877, Ensenada, B.C., México
3 Instituto de Astrofísica de Canarias, Tenerife, Spain
Accepted: 27 June 2008
Aims. SYMPA is the first instrument dedicated to the observation of free oscillations of Jupiter. Its principles and theoretical performance have been presented in Paper I. This second paper describes the data processing method, the real instrumental performance and presents the first results of a Jovian observation run, lead in 2005 at Teide Observatory.
Methods. SYMPA is a Fourier transform spectrometer which works at a fixed optical path difference. It produces Doppler shift maps of the observed object. The velocity amplitude of Jupiter's oscillations is expected to be below 60 cm s-1.
Results. Despite light technical defects, the instrument was demonstrated to work correctly, being limited only by photon noise. A noise level of about 12 cm s-1 was reached on a 10-night observation run, with 21% duty cycle, which is 5 time better than similar previous observations. However, no signal from Jupiter is clearly highlighted.
Key words: Sun: helioseismology / instrumentation: interferometers / methods: observational / techniques: spectroscopic / planets and satellites: individual: Jupiter / planets and satellites: formation
© ESO, 2008
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.