Volume 481, Number 3, April III 2008
|Page(s)||913 - 918|
|Published online||18 February 2008|
Optical spectropolarimetry with incomplete data sets
Department of Astronomy and McDonald Observatory, The University of Texas at Austin, 1 University Station, C1400, Austin, Texas 78712-0259, USA e-mail: firstname.lastname@example.org
Accepted: 29 January 2008
Context. Linear spectropolarimetry is a “photon-hungry” observing technique, requiring a specific sequence of observations to determine the Stokes Q and U parameters. For dual-beam spectropolarimeters, the Q and U Stokes parameters can be ideally determined using observations at retarder plate positions. The additional polarization signal introduced by instrumental effects requires the redundancy of observations to correct for these effects and to accurately measure the linear polarization of astronomical objects.
Aims. We wish to determine if the “instrumental signature corrections” for the Stokes Q and U parameters, and , are identical for observations with dual-beam spectropolarimeters. For instances when observations were only acquired at retarder plate angles, we wish to determine if the complete measurement of one Stokes parameter and the associated instrumental signature correction can be used to determine the other Stokes parameter.
Methods. We constructed analytical and Monte Carlo models of a general dual-beam spectropolarimeter to study the factors affecting the assumption and the uncertainty thereon. We compared these models with VLT FORS1 linear spectropolarimetry observations.
Results. We find that, in general, , with the variance around zero () being directly related to the signal-to-noise ratio of the observations. Observations of a polarized standard star, observed under identical instrumental conditions over the period of 2002-2007, show that the assumption of is generally true over a long period, although the absolute values of and vary between observational epochs. While the variance of is not dependent on the polarization angle, significant deviations from arise when .
Conclusions. Incomplete VLT FORS1 spectropolarimetry datasets, for which observations at only retarder plate position angles have been acquired, can be analyzed under the assumption that . The uncertainty associated with this assumption is directly related to the signal-to-noise ratio of the observations. This property of the analysis of spectropolarimetry, with dual beam spectropolarimeters, can be used to test for the presence of artifacts affecting individual observations and to assess the quality of the data reduction, when observations at all four retarder plate angles have been acquired.
Key words: instrumentation: polarimeters / polarization
© ESO, 2008
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.