Volume 442, Number 1, October IV 2005
|Page(s)||385 - 395|
|Published online||30 September 2005|
Global solar Doppler velocity determination with the GOLF/SoHO instrument
DSM/DAPNIA/Service d'Astrophysique, CEA/Saclay, 91191 Gif-sur-Yvette, France e-mail: firstname.lastname@example.org
2 AIM, Unité Mixte de Recherche CEA, CNRS, Université Paris VII, UMR N° 7158, France
3 Institut d'Astrophysique Spatiale, CNRS/Université Paris XI, 91405 Orsay, France
4 Observatoire de l'Université Bordeaux 1, BP 89, 33270 Floirac, France
5 Department of Physics & Astronomy, UCLA, Los Angeles, CA 90095-1562, USA
6 Instituto de Astrofísica de Canarias, 38205, La Laguna, Tenerife, Spain
7 Observatoire de la Côte d'Azur, Lab. Cassini, CNRS URA1362, 06304 Nice, France
8 Departamento de Astrofísica, Universidad de La Laguna, La Laguna, Tenerife, Spain
Accepted: 5 July 2005
The Global Oscillation at Low Frequencies (GOLF) experiment is a resonant scattering spectrophotometer on board the Solar and Heliospheric Observatory (SoHO) mission, originally designed to measure the disk-integrated solar oscillations of the Sun. This instrument was designed in a relative photometric mode involving both wings of the neutral sodium doublet (D1 at λ 5896 and D2 at λ 5890 Å). However, a “one-wing” photometric mode has been selected to ensure 100% continuity in the measurements after a problem in the polarization mechanisms. Thus the velocity is obtained from only two points on the same wing of the lines. This operating configuration imposes tighter constraints on the stability of the instrument with a higher sensitivity to instrumental variations. In this paper we discuss the evolution of the instrument during the last 8 years in space and the corrections applied to the measured counting rates due to known instrumental effects. We also describe a scaling procedure to obtain the variation of the Doppler velocity based on our knowledge of the sodium profile slope and we compare it to previous velocity estimations.
Key words: Sun: helioseismology / instrumentation: detectors
© ESO, 2005
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.