Volume 420, Number 2, June III 2004
|Page(s)||437 - 445|
|Section||Cosmology (including clusters of galaxies)|
|Published online||28 May 2004|
Antenna instrumental polarization and its effects on E- and B-modes for CMBP observations
IASF/CNR Bologna, via Gobetti 101, 40129 Bologna, Italy
2 IEIIT/CNR Torino, C.so Duca degli Abruzzi 24, 10129 Torino, Italy
Corresponding author: E. Carretti, email@example.com
Accepted: 18 March 2004
We analyze the instrumental polarization generated by the antenna system (optics and feed horn) due to the unpolarized sky emission. Our equations show that it is given by the convolution of the unpolarized emission map with a sort of instrumental polarization beam Π defined by the co- and cross-polar patterns of the antenna. This result is general, it can be applied to all antenna systems and is valid for all schemes to detect polarization, like correlation and differential polarimeters. The axisymmetric case is attractive: it generates an E-mode–like Π pattern, the contamination does not depend on the scanning strategy and the instrumental polarization map does not have B-mode contamination, making axisymmetric systems suitable to detect the faint B-mode signal of the Cosmic Microwave Background Polarization. The E-mode of the contamination only affects the FWHM scales leaving the larger ones significantly cleaner. Our analysis is also applied to the SPOrt experiment where we find that the contamination of the E-mode is negligible in the -range of interest for CMBP large angular scale investigations (multipole ).
Key words: polarization / cosmology: cosmic microwave background / instrumentation: polarimeters / methods: data analysis
© ESO, 2004
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.