Volume 386, Number 3, May II 2002
|Page(s)||1153 - 1156|
|Section||Numerical methods and codes|
|Published online||15 May 2002|
Observing programs of the Kyiv meridian axial circle equipped with a CCD micrometer
Astronomical Observatory of the Kyiv National University, Observatornaya 3, 04053 Kyiv-53, Ukraine e-mail: email@example.com
2 Main Astronomical Observatory, National Academy of Sciencies of Ukraine, Zabolotnogo 27, 03680 Kyiv-127, Ukraine
3 Mykolaiv Astronomical Observatory, Observatornaya 1, 54030 Mykolaev-MSP, Ukraine e-mail: firstname.lastname@example.org
Corresponding author: P. Lazorenko, email@example.com
Accepted: 8 February 2002
The present status of the Kyiv meridian axial circle after its refurbishment is described. The mounting of a new micrometer with the front-illuminated CCD 10401160 allows to observe stars up to mag. For the stars of 11–14 mag, the internal positional accuracy of a single observation is 0.05–0.10'', and stellar V values are determined with an accuracy of 0.03–0.06 mag. The project of observations of star fields in the direction of 209 extragalactic radio-sources in declination zone 0+30 is now in progress. The catalogue of positions, magnitudes and proper motions of secondary reference stars in regions with ICRF sources is expected to be compiled by 2003. Another project under way is a long-term astrometric survey of the equatorial zone to extend the Hipparcos-Tycho frame to fainter stars, and to measure their magnitudes and colours.
Key words: instrumentation: miscellaneous / astrometry: reference systems
© ESO, 2002
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.