Volume 365, Number 2, January 2001
|Page(s)||314 - 323|
|Section||Numerical methods and codes|
|Published online||15 January 2001|
Piston control with adaptive optics in stellar interferometry
Application to the GI2T interferometer and bimorph mirrors
Observatoire de la Côte d'Azur (OCA), Département Fresnel, 2130, route de l'Observatoire, 06460 Saint-Vallier-de-Thiey, France
2 Office National d'Études et de Recherches Aérospatiales (ONERA), DOTA, BP 72, 92322 Châtillon Cedex, France
Accepted: 18 October 2000
The general purpose of an adaptive optics system is to correct for the wavefront corrugations due to atmospheric turbulence. When applied to a stellar interferometer, care must be taken in the control of the mean optical path length, commonly called differential piston. This paper defines a general formalism for the piston control of a deformable mirror in the linear regime. It is shown that the usual filtering of the piston mode in the command space is not sufficient, mostly in the case of a bimorph mirror. Another algorithm is proposed to cancel in the command space the piston produced in the pupil space. This analysis is confirmed by simulations in the case of the GI2T interferometer located on Plateau de Calern, France. The contrast of the interference fringes is severely reduced in the case of a classical wavefront correction, even in short exposures, but is negligible with our algorithm, assuming a realistic calibration of the mirror. For this purpose, a simple concept for the calibration of the piston induced by a deformable mirror is proposed.
Key words: instrumentation: interferometers -instrumentation: adaptive optics - atmospheric effects -methods: numerical -techniques: interferometric
© ESO, 2001
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.