DFMS ion optics: neutrals enter the source where they are ionized (bottom center) before they pass the electrostatic analyzer (left bottom), the magnet (top left), and then arrive at the detector (top right). Taken from Balsiger et al. (2007). The inset shows a picture of the DFMS sensor in the same orientation.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.